Scott Volumeter USP 616 Ph.Eur. 2.9.34-1 Bulk Density Tester ISO3923/2
This apparatus is used to check the structural unifomity of diverse powdery sample materials inclusive metal powders by measuring the bulk density. Therefor the sample material is filled into a funnel containing a sieve with mesh size 1,00 mm (mesh 18). Below the funnel a baffle box with 4 glass baffles is mounted. The baffle box ends in a funnel that collects the sample material and let it run into a 25 ml measuring beaker. The bulk density is obtainable by difference weighing. Complete apparatus made of stainless steel including stable and non-slip stand. The modular build up of the complete apparatus especially of the baffle box allows an easy operation and cleaning. Normative Reference: • USP <616> bulk density and tapped density of powders, method II • European Pharmacopoeia 7.0 2.9.34. bulk density and tapped density of powders • Determination of apparent density Scott volumeter method DIN ISO 3923/2 • ASTM 3299 standard test method for apparent density of metal powders and compounds. Consisting of: • bulk funnel with sieve mesh size 1 mm (mesh 18), stainless steel • baffle box with glass baffles • measuring beaker 25 ml +/- 0,05, stainless steel • collection container, ’Clean Save’ laboratory glass • tripod with mark, stainless steel • wiper blade • open-jaw wrench • manual Accessoiries: • sieve mesh size 2,00 mm (mesh 10) • stainless steel tray • calibration certificate for measuring beaker • large tub, stainless steel Sale only to companies.